(1914 total: showing 1-20)
For the standard (non-NIST) version of the CS240 sensors, how does Campbell Scientific verify the ability of the sensors to measure and that the measurements made are calibrated to provide correct measurements?
How can data from LoggerNet be added into the SCADA software OSIsoft PI?
Why is NIST traceability beneficial for surface temperature sensors, such as the CS240, the 110PV, and the CS240DM?
How does the CH201 compare with the 18529 Morningstar SunSaver-10 Regulator?
Can the CH201 be connected to two 10 A sources, or does the total from the two sources have to be equivalent to 10 A?
If a CH201 is connected to both a solar panel and an AC/DC power convert, which power supply does the device use?
How quickly can measurements be attained using the ClimaVUE™50?
Is the calibration for the CS240DM NIST traceable?
Are the traditional methods of using RTD to measure the back-of-module temperature not used with the CS240 and CS240DM because these methods may cause hot spots?
Can multiple instances of Campbell Scientific SURVEYOR be run on a PC at the same time?
How tight do the wing-nuts on the CM226 integrated mounting stand need to be tightened to secure the solar sensor?
Can the SG000 determine the exact distance to a lightning strike?
How do I cancel a subscription for cellular data services?
Can I suspend my subscription for cellular data services?
Can I order more than a one-year subscription of cellular data services?
Can a cellular data services subscription be transferred to another customer?
How do I renew my cellular data services subscription?
What happens if my data use goes over the monthly allowance for my cellular data services plan size?
If I have a cellular data services plan, where can I find modem status information?
When selecting a recalibration service for a pyranometer, how do the services differ and what is the importance of ISO 9060?