我们的ET107 ETo气象站含有气象传感器、一根3米长的铝杆，以及安放CR1000M模块和7-Ahr密封可充电电池的环境机箱。即可采用交流适配电源，也可以用10 W太阳能板给电池充电。密封的圆形快速接头配在机箱的外面，实现了传感器快速连接。
The ET107 has components that have their own specifications.
To review these specifications, refer to the appropriate products below:
Replacement sensors for the ET107 have a -ET or -ETM suffix. These sensors are listed under Ordering.
The ET107 can also measure the:
One default program is for an ET107 with the 034B wind sensor and another default program is for an ET107 with a WindSonic1.
The ETsz() instruction requires the user to input lattitude, longitude, elevation, and crop type for a station's location in order to generate accurate calculations of ET. The Generic_ET_107_ETsz.CR1 and Generic_ET_107_ETsz_Windsonic.CR1 programs use the ConstTable() intruction to create a constants table that allows the user to easily input these values using Loggernet, PC400, or a keyboard display. These values only need to be set once, during station set up.
Calsense Command Center software can interface with Campbell Scientific’s VisualWeather software, which can retrieve values from the ET107. In the Command Center software, setting up a station involves connecting to the VisualWeather LoggerNet server and selecting variables from the weather station’s listing for wind speed, rainfall, and ETo.
Yes. However, special data logger programming is required. Contact an application engineer at Campbell Scientific for assistance.
The most significant difference is that the retired ET106 used the CR10X datalogger, whereas the ET107 uses the CR1000 datalogger. Other differences are the following:
The PS100 power supply that’s in the ET107 specifies a charging voltage of 15 to 28 Vdc or 18 Vac RMS. A 24 Vac transformer will blow the protective circuitry in the PS100.
No. VisualWeather must be purchased separately.
The communications options include direct serial connection, short haul modem, analog phone modem, cellular phone modem, radio, and Wi-Fi. Also, with an enclosure alteration, an Ethernet port can be added.
Note: Some of the listed telemetry options can drastically alter the power budget of the station.