The CS301 uses a silicon photovoltaic detector mounted in a cosine-corrected head to provide solar radiation measurements. Its dome-shaped head prevents water from accumulating on the sensor head. To eliminate internal condensation, the sensor head is potted solid and the cable is shielded with a rugged Santoprene casing. The CS301 is calibrated against a Kipp & Zonen CM21 thermopile pyranometer to accurately measure sun plus sky radiation.
|Light Spectrum Waveband||360 to 1120 nm (wavelengths where response is 10% of maximum)|
|Measurement Range||0 to 2000 W/m2 (full sunlight ≈1000 W/m2)|
|Absolute Accuracy||±5% for daily total radiation|
|Calibration Factor||5 W/m2/mV|
|Cosine Correction Error||±5% at 75° zenith angle; ±2% at 45° zenith angle|
|Temperature Response||0.04 ± 0.04% per °C|
|Response Time||< 1 ms|
|Long-Term Stability||< 2% per year|
|Operating Temperature Range||-40° to +70°C|
|Relative Humidity Range||0 to 100%|
|Output Sensitivity||0.2 mV/W/m2|
|Diameter||2.4 cm (0.9 in.)|
|Height||2.5 cm (1.0 in.)|
|Weight||65 g (2.3 oz) with 2-m (6.6-ft) lead wire|
Please note: The following shows notable compatibility information. It is not a comprehensive list of all compatible products.
Accurate measurements require the sensor to be leveled using a 18356 leveling fixture. This leveling fixture incorporates a bubble level and three adjusting screws. The 18356 mounts to a crossarm or a tripod or tower mast using the CM225 mounting stand. The CS301 should be mounted away from all obstructions and reflective surfaces that might adversely effect the measurement.
One differential analog input channel per probe is required.