该型号机箱与ENC10/12 类似，但其机箱门更高，这允许了更高的外设安装在ENC10/12R内部。ENC10/12R 通常用于安装一个较小的数据采集器、供电电源和一个小型外设。还可以用于安装远距离的外设，例如扩展板。
The ENC10/12R's backplate is prepunched with half-inch-on-center holes suitable for attaching a data logger, power supply, and a communications or measurement and control peripheral.
The ENC10/12R is shipped with the 7363 enclosure supply kit that consists of desiccant, a humidity indicator card, cable ties, wire tie tabs, putty, grommets, screws, and PVC coupling. Additionally, Campbell Scientific offers a CS210 Enclosure Humidity Sensor for monitoring relative humidity inside of the enclosure. (See Ordering Info on the web page.)
|Construction||Fiberglass-reinforced polyester enclosure with door gasket, external grounding lug, stainless-steel hinge, and lockable hasps|
|Color||White (Reflects solar radiation, reducing temperature gradients inside the enclosure without using a separate radiation shield.)|
|Enclosure Classification||NEMA 4X (before being modified for cable entry, conduit, or ground lug)|
|Number of Cable-Entry Seals||1 medium, 2 small|
|Weight||4.1 kg (9 lb)|
Please note: The following shows notable compatibility information. It is not a comprehensive list of all compatible products.
|CR3000 (retired)||The ENC10/12R is compatible with the CR3000. Similar to the CR800/CR850 and CR1000, the CR3000 must be mounted vertically rather than horizontally. No space remains in the enclosure for additional peripherals.|
|CR800||Although the CR800 is compatible, this data logger can only fit inside the ENC10/12R if it is mounted vertically rather than horizontally.|
|CR850||Although the CR850 is compatible, this data logger can only fit inside the ENC10/12R if it is mounted vertically rather than horizontally.|
Retired peripherals that won’t fit in the same ENC10/12R enclosure as the data logger/power supply include the AVW4, AM416, COM100, CR10KD, DC1765, and RC35.
If you purchased a Stahlin enclosure from Campbell Scientific and it has a failed gasket, we can help you. We will need you to provide an image of the failed gasket, verify the size of your enclosure, and provide your contact information. We can then arrange to have a replacement gasket sent to you from Stahlin. For assistance, contact Technical Support at 435.227.9100.
The following information is applicable to many of Campbell Scientific’s standard enclosures and prewired enclosures.
Note: The procedure below will not work for torn or damaged seals.
To make a field repair to a loose gasket on an enclosure, first acquire an adhesive, such as standard Gorilla Glue, that is rated for use with polyurethane foam. The adhesive should be compatible with, and recommended for, use with polyurethane foam. Follow these steps: